Difference between revisions of "Electronics notes/Temperature sensing"
m (→DHT11, DHT22)
|(One intermediate revision by the same user not shown)|
|Line 83:||Line 83:|
Revision as of 13:53, 24 March 2020
|This article/section is a stub — probably a pile of half-sorted notes, is not well-checked so may have incorrect bits. (Feel free to ignore, fix, or tell me)|
Inexpensive High range of temperature Not highly accurate - assume you won't get better than 1 degree Celcius of resolution, less if you don't calibrate well
Age with time and high-temperature use, so will need occasional recalibration and/or replacement.
They produce voltage due to the thermoelectric effect - on the order of millivolts so most uses, and accurate use, requires amplification (with high input impedance to avoid the measurement affecting the thermocouple itself).
Most resistors vary their resistance with temperature.
A thermistor (thermal resistor) does it intentionally, and more pronounced.
NTC: negative temperature coefficient, resistance drops (logarithmically) as its body temperature increases
PTC: positive temperature coefficient,
The 'at-rest' resistance varies with intent
They are frequently used in temperature sensing, temperature regulation, and (over)current protection.
Perhaps the he simplest way to get a voltage from a thermistor (think ADC, comparator) is to have it be one leg of a voltage divider.
A power thermistor is a very low-resistance (NTC) thermistor in series with your main current, as a current limiter and/or (self-resetting) overcurrent protector.
One use is to spread the inrush current (in transformers and such) over more time:
- place in series with the primarily coil
- when cold (just switched on) it typically has a few hundred ohm resistance
- and once it warms it (few seconds later) goes to under an ohm.
This lessens the magnitude of the sudden current that can happen right after you switch something on.
The voltage across a diode will decrease by approx 2 mV per °C in a fairly linear way.
You need amplification and probably a DAC, accuracy isn't great, but it's very convenient inside FPGAs, processors, and such.
- Temperature (NTC thermistor)
- 0-50 ℃
- output reported to 1℃ but don't assume more than ±2℃ accuracy
- Relative Humidity
- Range: 20-90 ％RH (note: range and accuracy vary somewhat with temperature)
- output reported to 1% but don't assume more than ±5％RH accuracy
In a quick test with two DHT11 modules next to each other, temperature was reported 0.0 to 1.0 degrees apart on average, but humidity was ~11 %RH apart on average (fairly consistently so you can probably improve this a little via calibration against a known value).
Pins: 3.5-5.5V, data, NC, Gnd. You will want a pullup resistor on the data pin, mostly because it's bidirectional onewireish(verify).
Power: ~0.5mA while measuring, 0.1mA idle
- RH: 0-100% range, 2-5% accuracy
- Temp: -40 to 80°C range, ±0.5°C accuracy